ADVIA 360 Hematology System
Compact footprint and remarkable accuracy in a cost-efficent system
Volumetric impedance method
22 parameters, including 3-part WBC differential: WBC, LYM, MID, GRA, LYM%, MID%, GRA%, RBC, MCV, HCT, HGB, MCH, MCHC, PLT, MPV, RDW-SD*, RDW-CV, PCT*, PDW-SD,* PDW-CV,* P-LCR,* P-LCC*
Long-term sample stability studies on blood specimens drawn in K2- and K3-EDTA collection tubes at room temperature showed no significant clinical variation for all parameters between 30 minutes and 7 hours post-phlebotomy. MPV results can show instability in the first 2 hours but are stable afterwards.
Three cyanide-free reagents (Diluent, Lyse, and Cleaner)
*Parameter is not available for measurement in the U.S. Due to regulatory reasons its future availability cannot be guaranteed.
The products/features (mentioned herein) are not commercially available in all countries. Due to regulatory reasons their future availability cannot be guaranteed. Please contact your local Siemens organization for further details.